The Internet of Things (IIoT) offers unprecedented possibilities for industries. Stay ahead with EcoStruxure™ Plant and Machine.Learn more
...with the EcoStruxure™ Facility Expert (formerly known as Facility Hero) app in hand! Discover how you can optimize your maintenance efforts today.Learn more
Access tailored services, 24/7 self-service and expert help. Anywhere, anytime, any project.Discover now
The story behind the Shedd Aquarium, a smart, sustainable facility that connects people to animals, with EcoStruxure™.Discover how
Welcome to the Schneider Electric Global Website×
A user would like to configure a KWh pulse output from the front screen of a ION7350
ION 7350 Front Screen Setup
Front panel configuration of the ION7350
To configure the pulse outputs from the front screen or panel of an ION7350:
1. From the home screen of the ION7350 press the circle button
2. Scoll down to Adv. Meter Setup, press the circle button
3. Scroll down to Calibration Pulser Module, press the circle button
4. Select KWh exp Pulse, press the circle button
5. The following options will be available for customization:
a. CAP2 Pulse Width = This defines the width of the pulse sent from the ION meter.
b. CAP2 KT = This specifies the number of Wh's per pulse in seconds.
c. CAP2 Int Mode = Specifies the integration
d. CAP2 Output Mode = Sets the type of output
e. CAP2 Port = Rear I/O port that will send the KWh pulse
Sign up for all the latest solutions, offers and best practices straight to your inbox.
We are excited that you have joined the group. You will receive your welcome email soon describing the program and what to expect in the upcoming weeks. Enjoy!