The Internet of Things (IIoT) offers unprecedented possibilities for industries. Stay ahead with EcoStruxure™ Plant and Machine.Learn more
...with the EcoStruxure™ Facility Expert (formerly known as Facility Hero) app in hand! Discover how you can optimize your maintenance efforts today.Learn more
Access tailored services, 24/7 self-service and expert help. Anywhere, anytime, any project.Discover now
The story behind the Shedd Aquarium, a smart, sustainable facility that connects people to animals, with EcoStruxure™.Discover how
Welcome to the Schneider Electric Global Website×
There are 2 different methods to determine the firmware level in a Conzerv EM6400/DM6200/PM1200 meter if there is no communications to the meter: The label on the back of the meter and from the front panel.
The firmware revision can be identified by looking at the label on the back of the meter next to "VER"
E.g. This model has a firmware revision of 03.03.05.
From the front panel, cycle either up or down from the root menu to "DIAG". Once the display reads DIAG, press right and then down to get to "DIA2". Diagnostic screen 2 provides the product model (first line) and firmware revision number (lines 2 and 3).
E.g. This is a EM6400 meter with a firmware revision of 03.02.12.00.
Sign up for all the latest solutions, offers and best practices straight to your inbox.
We are excited that you have joined the group. You will receive your welcome email soon describing the program and what to expect in the upcoming weeks. Enjoy!