Here’s what more than 2,500 business decision-makers around the world say.Download the report
...with the Facility Hero app in hand! Discover how you can optimize your maintenance efforts today.Learn more
Access tailored services, 24/7 self-service and expert help. Anywhere, anytime, any project.Discover now
Everything you need to know about our IoT technology backbone.Learn more
Welcome to the Schneider Electric Global Website×
The PM800 requires a ferrite kit
The higher frequencies in the control power wire can produce radio waves which put the PM8 is a class of devices called "unintentional radiators". Though unintentional, since it is a radiator, the device must meet certain FCC requirements so it does not interfere with other radiating devices.
The PM8 ferrite kit has been supplied to make the device compliant with FCC part 15 class A EN 55011.
Unintentional radiators fall under two categories: Class A [intended for use in business/industrial/commercial environments] and Class B [intended for use in or adjacent to residential/small office environments]. The PM8 falls under Class A, which is a less stringent requirement than Class B.
Not all PM8's need the kit. Newer hardware versions in the PM8 have eliminated the need for a ferrite core. Refer to the Ferrite Kit Installation Guide to determine which PM's need the ferrite core.
The ferrite core should be installed on all models that require it to meet FCC requirements.