The Internet of Things (IIoT) offers unprecedented possibilities for industries. Stay ahead with EcoStruxure™ Plant and Machine.Learn more
...with the EcoStruxure™ Facility Expert (formerly known as Facility Hero) app in hand! Discover how you can optimize your maintenance efforts today.Learn more
Access tailored services, 24/7 self-service and expert help. Anywhere, anytime, any project.Discover now
The story behind the Shedd Aquarium, a smart, sustainable facility that connects people to animals, with EcoStruxure™.Discover how
Welcome to the Schneider Electric Global Website×
Goals and Symptoms
When sending the .dcf meter template file into the meter using ION Setup, it will return an error message and say "Unable to download dcf(xml) file to a non-RMP device"
Causes and Fixes
Rapid Meter Programming (RMP) is not supported in ION7500RTU meters, hence the .dcf files can not be sent to the meter using ION Setup. Instead, simply paste the .fwn version of the meter template file using ION Setup. ION Setup will treat this fwn file the same as the .dcf file with meters that support RMP. Please see the template file attached.
(Removed File URL: 198385_B79E/7500-RTU_V1.1.0.fwn)(Removed Image URL: /PubResEXPORT.nsf/2b87ee90be777fc085257c28006ee4ef/f23cc1f31e56d096c12575a70052a263/fl_block_5/0.56E?OpenElement&FieldElemFormat=gif)
(Removed Image URL: /PubResEXPORT.nsf/2b87ee90be777fc085257c28006ee4ef/f23cc1f31e56d096c12575a70052a263/fl_block_5/0.26EE?OpenElement&FieldElemFormat=gif)
Date Created: March 3, 2009
Last Revised: April 29, 2009
Original Author: KG
All content © 1992-2009 Schneider Electric
Sign up for all the latest solutions, offers and best practices straight to your inbox.
We are excited that you have joined the group. You will receive your welcome email soon describing the program and what to expect in the upcoming weeks. Enjoy!